ped-lock Seal pots are used in the flow measurement of high-density and corrosive media to protect differential pressure measuring instruments from these media. The seal pots are filled with an unmixable sealing liquid which, due to the difference in density, forms a sealing layer towards the medium
The primary use for the ped-lock range of condensate pots is to increase the accuracy of flow measurement in steam pipelines. They provide an interface between the vapour phase and the condensed phase in the impulse lines. The condensate pots are available in a range of materials and have been designed in accordance with ASME VIII Div 1 and produced in an ASME coded workshop. All condensate pots are CE* marked to PED 2014/68/EU for use with Group 2 Gases. Installation can be either vertical or horizontal lines between the primary (Flow Meter) and the secondary (transmitter/ gauge) to act as a barrier to the line fluid permitting direct sensing of the flow conditions. Units should be mounted at the same level minimising possible error that could arise due to unequal head of fluid in the connecting pressure lines. Please see our Condensate Pot Installation Guide.
Item Code: 3582
Port of Dispatch: ahmedabad airport
Production Capacity: 10000pcs
Delivery Time: 1week
Packaging Details: INSTALLATION TIPS
1. Use stainless steel tubing for the connection to the
flanges or steam line.
2. Place the pot nearer the steam line or flange
3. Keep both the condenser and condensate chamber
at the same level as the upper connection point. (An
unequal level will cause significant error due to false
4. If the flow device (in the steam line) is installed in a
vertical pipe, align the lower connection point to the
steam pipe with the upper connection point.